- Technical report
- Open Access
A grooved glass surface-plate for making a flat polished surface
© The Author(s) 2017
- Received: 14 September 2016
- Accepted: 8 December 2016
- Published: 3 January 2017
To obtain a flat polished surface for microchemical analyses such as EPMA, SIMS, and ATR micro-FTIR, a glass plate with a grooved surface was developed for hand polishing with an abrasive film. It eases the polishing process by minimizing slipping or sticking, and results in negligible relief in the sample surface.
- ATR micro-FTIR
- Polished sections
The flatness of the polished surface is important in the preparation of samples for microanalysis by techniques such as EPMA, SIMS, and ATR micro-FTIR, which are essential techniques in the quantitative analysis of geological samples. Conventional polishing techniques involve final buffing with a cloth, which leads to undesired relief that develops on the polished surface near the boundaries of relatively soft and hard materials, for example soft glass inclusions in hard quartz phenocrysts and hard glassy ash particles mounted in soft resin. While the unevenness is essential for the examination, identification, and textural interpretation of ore minerals using the reflected light microscope (Craig et al. 1981), such unevenness may interfere with successful analysis with either micro-FTIR spectroscopy (Lowenstern and Pitcher 2013) or the electron microprobe (e.g., Lengauer et al. 1997). The method of hand polishing described here uses a newly developed glass plate with a grooved surface that greatly aids in achievement of a flat polished surface.
The grooves on a glass surface can be made by etching using dilute hydrofluoric acid on a masked float glass plate by a trained chemist. Shallow grooves can also be drilled into a hard metal such as bronze, in lieu of using glass. Plates with the listed specification are also available commercially from a Japanese dealer (Tsukuba-hikari-kagaku Co, Ltd; Product No. TH-025; FAX: +81-29-851-4696; WWW: http://t-h-k.jp).
I thank Dr. N. Hoang for help in making a prototype surface-plate, Dr. J. B. Lowenstern for encouragement and for reading an early draft of the manuscript, two anonymous reviewers for their constructive comments, and Prof. Nakagawa for handling this paper.
The present study was supported by the National Institute of Advanced Industrial Science and Technology (AIST) of Japan, who holds a patent relating to the content of the manuscript. The author has no competing financial interests or personal interests which may have impacted the interpretation of the presentation of this information. The authors declare that they have no competing interests.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Craig JR, Vaughan DJ, Hagni RD (1981) The preparation of samples for ore microscopy. Ore microscopy and ore petrography, Wiley New York. ISBN: 0-471-55175-9Google Scholar
- Lengauer W, Bauer J, Bohn M, Wiesenberger H, Ettmayer P (1997) Electron-probe microanalysis of light elements in multiphase diffusion couples. Mikrochim Acta 126:279–288View ArticleGoogle Scholar
- Lowenstern JB, Pitcher BW (2013) Analysis of H\(_2\)O in silicate glass using attenuated total reflectance (ATR) micro-FTIR spectroscopy. Am Miner 98:1660–1668View ArticleGoogle Scholar