- Open Access
X-ray computerized tomography analysis and density estimation using a sediment core from the Challenger Mound area in the Porcupine Seabight, off Western Ireland
© The Society of Geomagnetism and Earth, Planetary and Space Sciences (SGEPSS); The Seismological Society of Japan; The Volcanological Society of Japan; The Geodetic Society of Japan; The Japanese Society for Planetary Sciences; TERRAPUB. 2011
- Received: 27 May 2009
- Accepted: 13 December 2010
- Published: 28 February 2011
X-ray computerized tomography (CT) analysis was used to image a half-round core sample of 50 cm long recovered from near Challenger Mound in the Porcupine Seabight, off western Ireland during the Integrated Ocean Drilling Program Expedition 307. This allowed three-dimensional examination of complex shapes of pebbles and ice-rafted debris in sedimentary sequences. X-ray CT analysis was also used for the determination of physical properties; a comparison between bulk density by the mass-volume method and estimated density based on linear attenuation coefficients of X-ray CT images provides insight into a spatially detailed and precise map of density variation in samples through the distribution of CT numbers.
- X-ray CT
- bulk density
- sediment core
- Integrated Ocean Drilling Program (IODP)
- Expedition 307
X-ray computerized tomography (CT) is a nondestructive technique that allows three-dimensional visualization of internal structures of samples, determined mainly by variations in their density and atomic composition. Since X-ray CT was developed as a medical imaging technique in the early 1970s (Hounsfield, 1973), it is increasingly used and has proven useful in the field of earth sciences (e.g., Mees et al., 2003). For example, it has been used to present quantitative aperture data for three-dimensional interconnected fracture networks (Pyrak-Nolte et al., 1997), to image the internal structure of fragile sea-floor hydrothermal vent samples (Tivey and Singh, 1997), and to provide information on the physical properties of sediment cores (e.g., Orsi et al., 1994). Through the Ocean Drilling Program (ODP) and the Integrated Ocean Drilling Program (IODP), X-ray CT system has gained acceptance as a routine core analysis tool, and some attempts have been made to evaluate X-ray CT imaging for observation and description of core samples (e.g., Ashi, 1997; Iturrino et al., 2004; Tanaka and Nakano, 2009). However, many studies were treated X-ray CT images only as a non-invasive imaging tool.
In this paper, three-dimensional imaging analysis of sediment core from the IODP Expedition 307 was performed by a medical X-ray CT system, and was compared and contrasted with line-scan digital core images and X-ray radiographs to highlight some characteristics to be considered when using these data. Also, the possibility and limitation of density estimates based on X-ray attenuation is considered.
2.1 Study area and core sample
Hole U1318A (51°26.16′N, 11°33.0′W; 423 m water depth), which is located upslope from the Belgica mound province including Challenger Mound, was drilled continuously to 142.2 mbsf (meters below sea floor) to constrain the stratigraphic framework of the slope and to capture the regional environment system, and 15 cores were taken with a recovery of 90.2%. A recovered core is divided into 1.5 m sections that are numbered serially from the top. These whole-core sections were scanned with the multisensor track (MST), included Gamma ray attenuation (GRA) bulk density measurements on board in May 2005. Core U1318A-5HSection 4 (1.5 m long) was split into halves, referred to as “archive-half” and “working-half”, on board in May 2005. Whole-core photographs of the “archive-halves” were taken on film during core description in May 2005, and subsequently line-scan digital images from the film at 3000 dpi were captured with the Digital Imaging System (http://iodp.tamu.edu/database/coreimages.html). These cores and scientific data collected by shipboard scientists were archived on the ODP/TAMU servers (ODP Information Technology and Data Services, 2007).
Half-round core sample from U1318A-5H-Section 4 was cut into 50-cm long section at the Bremen Core Repository in October 2005. This used half-round core of a diameter of 66 mm was covered with plastic shrink-wrap to avoid dry out. The allocated section 307-U1318A-5H-4, 22–72 cm (42.92–43.43 mbsf) from the upper Pleistocene clay layer, was used to know the stratigraphic sequences. This section comprises dark grayish brown to very dark gray silty clays with carbonate contents of about 15 wt% (Ferdelman et al., 2006). Dropstones are much less frequent and lamination occurs on a centimeter to millimeter scale. Titschack et al. (2009) showed that the average sedimentation rate of this section is about 7.1 cm/ka. Assuming a constant sedimentation rate, this core sample records at intervals of about seven thousands years.
2.2 X-Ray computerized tomography measurements
A medical X-ray CT scanner (Hitachi CT-W2000) of the Geological Survey of Japan, AIST, was used to acquire the CT images. A medical X-ray CT system such as used in this study was originally designed for use on human subjects to image soft tissue and bone. Cores used in this study consist mainly of sediments with densities of ~2 g/cm3, which is consistent with that of human bone. It might be reasonable to use medical X-ray for internal structures of wet sediments.
All samples were scanned with X-ray peak energy of 120 kV with 150-mA current, with an in-plane resolution of 0.313 × 0.313 mm2/pixel and the slice thickness of 1 mm. Slice acquisition times were 4 seconds per scan and a 160-mm field of reconstruction was captured in a 512 pixel × 512 pixel image. 545 contiguous two-dimensional 16-bit CT images at intervals of 1 mm sequentially from core top to bottom perpendicular to the longitudinal axis of the core were acquired in October 2005. These data were used to reconstruct the three-dimensional images. After image reconstruction, relative values of linear attenuation coefficient are stored for each voxel.
The intensity of the transmitted X-ray beam is usually expressed as CT number, Hounsfield unit (HU). The HU scale is a linear transformation of the original linear attenuation coefficient measurement into one where water is assigned a value of zero and air is assigned a value of .1000. When μw and μ are the linear attenuation coefficients of water and a substance of interest, the CT number is defined as . Thus, a change of one CT number corresponds to 0.1% of the attenuation coefficient difference between water and air. The density is the predominant factor determining X-ray attenuation at the energies of 120 kV used in this study, which is used for most medical scanners (McCullough, 1975). Therefore, each CT number represents a 0.1% changes, or approximately 0.001 g/cm3, change in density with respect to calibration density scale.
2.3 X-ray radiograms
X-ray radiography is based on the differential transmission of radiation through a sample. This technique is nondestructive, and provides density variations within a sample. To obtain a better image, the half-round core sample was cut and divided into slabs with dimensions of 200 × 30 × 7 mm; with z, x, y directions in Fig. 3(a), respectively. Three 7 mm-thick slabs, s1, s2, and s3, along the direction from core top to core bottom, are prepared to accentuate sedimentary structures and to image ice-rafted debris (IRD). s1 and s3 are cut out from the central part of the half-round core sample from 7 mm up to 14 mm in depth intervals, however s2 is cut from the core surface in depth intervals of up to 7 mm, avoiding the pebble. X-ray radiographs of slab samples were taken using a SOFRON STA- 10050 X-ray unit at the Geological Survey of Japan, AIST, in November 2005, using a voltage of 40 kVp, a current of 3.5 mA and an exposure time of 40 seconds. X-ray radiographs were directly exposed on FUJIBROWP FM3 photographic paper. Dark/light color in image indicates low/high density (high/low X-ray transmission), respectively.
2.4 Density measurements
The shipboard GRA bulk density measurements were carried out on the unsplit core with sampling interval and measurement width of 5 cm and ∼5 mm, immediately after the core liner is cut into sections in May 2005 (Ferdelman et al., 2006). This measurement is based on the principle that the 137Cs attenuation, mainly by Compton scattering, of a collimated beam of gamma rays passing through a known volume of sediment is related to material density (Boyce, 1976). The obtained GRA bulk density in this core section ranges from 1.98 to 2.04 g/cm3.
Density measurements by Pycnometer (MAD density) using discrete core specimens of ∼8 cm3 (Blum, 1997) were conducted after splitting core in May 2005. The MAD density is 1.877 g/cm3 at 42.90 mbsf (Ferdelman et al., 2006), which is in good agreement with the GRA density. This value is the direct estimate of the average bulk density, and can be used to calibrate the GRA bulk density data.
Bulk density measurements by the mass-volume method were also carried out at the Geological Survey of Japan, AIST, in June 2008. Discrete samples from the lithified host sediments of about 7 cm3 were selected using plastic cubes at 5 cm intervals from as close to the center of the core as possible to measure density. Obtained wet-bulk density was determined to be 1.65 to 1.86 g/cm3. On the other hand, the density of the sedimentary pebble sample at about 60 cm was determined to be 2.74±0.04 g/cm3 using the water saturation method at the Geological Survey of Japan, AIST, in August 2007 (Furukawa, personal comm.).
X-ray CT images on y-z and z-x planes as defined in Fig. 3(a)x are shown in Fig. 3(b), although image reconstructions in any directions are possible after the data acquisition. In Fig. 3(b)x, X-ray CT-generated grayscale images were constructed using histogram equalization with the lower limit value of water, which provides contrastive images for structure observations. Left and right sides of Fig. 3(b) show cross sections on y-z and z-x planes every 15 and 12 units of in-plane resolution (0.313 mm), respectively. The core is not perfectly aligned to the center of stage of X-ray CT scanner; therefore the core appears tilted in the images. Two holes at 36.1 and 74.0 cm, due to shear strengths measurements with a hand held torvane sampler on board, are clearly visible in both y-z and z-x planes. Curved laminar structures possibly due to drilling and sampling processes, are observed near the linear. Significant differences in CT numbers among sediments, IRD (some examples are highlighted by white circles on y = 72 plane in Fig. 3(b)), and pebble at 60 cm are represented by grayscale. Although CT numbers of sediments are similar (clearly shown in Fig. 2), laminae appear in CT number variations by the enhanced images as shown in Fig. 3(b), implying that these structures retain density contrasts. Figure 3(c) shows the CT number variations along the black line on y = 72 plane in Fig. 3(b). CT number difference between peak and trough values are approximately 100–200 as in Fig. 3(c), and it corresponds to 0.1–0.2 g/cm3 changes in density, large enough to discriminate density contrasts. Using the core sedimentation rate of 7.1 cm/ka (Titschack et al., 2009), CT number variations are assessed with temporal resolution of the order often years.
Three kinds of images as shown in Fig. 4 are compared with each other to highlight their characteristics, advantages, and limitations. Line-scan digital core images provide detailed visual information on the surface. X-ray radiograph system produces 2-D images of internal 3-D structures, but in a single 2-D shadow projection the depth information is completely mixed. X-ray radiographs reveal laminae on a millimeter scale more clearly but nonquantitatively, and highlight many IRDs as bright spots against dark-colored background sediments, partly because they reflect an integration of information over the slab thickness of 7 mm. On the other hand, X-ray CT system allows us to visualise and measure complete 3-D object structures without sample preparation, and X-ray CT image can be sliced in any sections and reveal sedimentary and post sedimentation structures and materials that cannot be seen on the split core surface. However, because of the lower resolution of X-ray CT images, smaller targets may not be imaged.
Variations in grayscale values of the line-scan digital core image, X-ray CT image and X-ray radiograph reflect laminae. Based on the line-scan digital core image, laminae curved downwards due to drilling disturbance close to liners were frequently observed, which are easily recognized by X-ray CT image. The X-ray CT image (Fig. 4(b)) also captures disturbance of laminae near the pebble at 60 cm. This pebble is clearly captured in the X-ray CT images but cannot be seen on the split core surface. However, disturbances of original sedimentary structures at corresponding range are evidenced in the line-scan digital core image as shown in Fig. 4(c).
Regions near the center of core excluding the pebble at 60 cm and two data gaps at 36.1 cm and 74.0 cm are chosen to show CT number variations in Fig. 5. CT numbers in each voxel of reconstructed CT image are plotted as a function of core section depth. Each average value at x = 150, 160, and 170, using 5 data at y = 80, 90, 100, 110, 120 is used to exclude outliers and to reflect a larger spatial area, as shown in gray lines in Fig. 5. Although the direction of core section depth is not exactly parallel to sediment layer stratification, each CT number variations and three average values show similar trends with peaks in phase. As mentioned before, there might be expected a sort of relation between variations of CT number and densities. Gray lines in Fig. 5 show density variations corresponding to distinct sedimentary sequence, which clearly shown in different colours in Fig. 4. Figure 5 also shows a moving average of 10 CT numbers for the core section depth using 15 profiles at (x, y) = ([150, 160, 170], [80, 90, 100, 110, 120]), to capture long-term trends.
Two GRA density peaks at depths of 35 and 60 cm are shown in Fig. 5, although GRA density changes are relatively small all through this sample. The peak at 35 cm is also recognized at the similar depth in the line-scan digital core image and X-ray radiograph as a bright-colored area in Figs. 4(a) and (c), which corresponds to higher CT number. At this cross section depth, density by mass/volume method is rather high. The GRA density has the second highest value at 60 cm through this sample, where averaged CT number also has a peak at this depth. On the other hand, CT number shows a high peak at 64 cm. This part coincides with a concentration of a white fine-sand layer, only 5–10 mm thick in the line-scan digital core image, X-ray CT image and X-ray radiograph as shown in Fig. 4. However, densities obtained from GRA measurements and mass/volume methods do not correspond to this high CT number. The peak at 64 cm may be out of the target area of GRA density measurements collected at 5 cm intervals with the measurement width of ∼5 mm.
Several attempts have been made to convert CT numbers to equivalent sediment bulk density and have verified the linear correlation between bulk density and X-ray attenuation (e.g., Orsi et al., 1994), although there are many uncontrollable variables, including machine-dependent parameters and statistical parameters of samples such as mean value and standard deviation of CT numbers. Previous calibration experiments on the Technicare Δ-100 CT scanner at 120 kV/25 mA using marine sediments from the northern Gulf of Mexico continental shelf, showed a linear response between bulk density (ρ) and CT number (HU) described by (Orsi et al., 1994). They estimated the average density resolution of the system much smaller than 0.01 g/cm3 with the bulk density range from 1.96 to 2.08 g/cm3. A calibration procedure by Orsi and Anderson (1999) using two types of material showed the difference of gradient between the curve for SiO2 and CaCO3 CT number versus bulk density plots, namely and over a range of 1.0 and 2.2 g/cm3. Inazaki et al. (1995) indicated that the accuracy of bulk density measurements estimated from the CT number is better than ±0.02 g/cm3 by calibration with known standards using medical X-ray CT system (TCT-700S) at 120 kV/55–200 mA. Their suggested linear regression is , using water glasses and pressed bentonites with bulk density ranges between 1.10 and 1.95 g/cm3.
Standard conventional techniques for bulk density analysis in sediment cores use raw data from GRA or mass/volume method. These density values obtained from samples represent the gross average. Neither have a spatial resolution better than X-ray CT data-derived density. Moreover, X-ray CT data-derived density offers advantages over these standard conventional techniques as it gives a threedimensional distribution anywhere in the sample. Also, because of the simple process of the conversion from CT number to density, it is expected to reduce the possibility of human error.
Three-dimensional quantitative analysis using X-ray CT combined with X-ray radiographs and line-scan digital core images of a marine sequence recovered during IODP Expedition 307 in the Porcupine Basin is an effective method for core characterization. This paper confirms that X-ray CT proves to be a relatively quick and accurate method for high-resolution and non-destructive analysis of three-dimensional internal structures of core samples without the special sample preparation. A medical CT system at relatively low resolution provides an image of the largescale features and allows correlation with core samples, and has generally been used due to their availability and relative ease of use. CT scanning just after coring and before splitting may be efficient to avoid a degradation of cores. Effective and efficient use of X-ray CT systems allows for the possibility for rapid systematic characterization of three-dimensional structural features, and may improve sub-sampling and core-processing procedures. Because X-ray attenuation is sensitive to density variations, X-ray CT systems also offer the possibility of quantitative density measurement. Its rapid and quantitative results can easily be used to map subtle detailed density variations in three-dimensions.
We appreciate the valuable and detailed comments of Dr. Dan Sinclair, an anonymous reviewer, and the editor. Their comments helped us to improve the manuscript and clarify our arguments. We thank the IODP and Transocean crews of the JOIDES Resolution of IODP Expedition 307, and the staff at the Bremen Core Repository. The core sample from Hole U1318A (307-U1316A-5H-4) was provided by IODP. We would like to thank Dr. Yoshito Nakashima for his technical assistance during X-ray CT data collection and Dr. Ryuta Furukawa for his density measurement of the pebble. Some of figures in this manuscript were prepared using GMT (Wessel and Smith, 1998)
- Ashi, J., Computed tomography scan image analysis of sediments, in Proc. ODP, Sci. Results, edited by Shipley, T. H., Y. Ogawa, P. Blum, and J. M. Bahr, 156, 151–159, 1997.Google Scholar
- Blum, P., Physical properties handbook: a guide to the shipboard measurement of physical properties of deep-sea cores, ODP Tech. Note, 26, 1997.Google Scholar
- Boyce, R. E., Definitions and laboratory techniques of compressional sound velocity parameters and wet water content, bulk density, and porosity parameters by gravimetric and gamma-ray attenuation techniques, in Initial Reports Deep Sea Drilling Project 33, edited by Schlanger, S. O., E. Jackson et al., 931–958, US Government Printing Office, Washington, DC, 1976.Google Scholar
- Ferdelman, T. G., A. Kano, T. Williams, J.-P. Henriet, and the IODP Expedition 307 Scientists, Modern carbonate mounds: Porcupine Drilling, in Proceedings of the Integrated Ocean Drilling Program, Volume 307: College Station, Texas, Integrated Ocean Drilling Program Management International, Inc., doi: 10.2204/iodp.proc.307.2006, 2006.Google Scholar
- Hounsfield, G. N., Computerized transverse axial scanning (tomography): Part I. Description of system, Br. J. Radiol., 46, 1016–1022, 1973.View ArticleGoogle Scholar
- Inazaki, T., Y. Inouchi, and T. Nakano, Use of medical X-ray CT scanner for nondestructive and quantitative analysis of lake sediments, Bull. Geol. Surv. Jpn., 46, 629–642, 1995 (in Japanese with English abstract).Google Scholar
- Iturrino, G. J., R. A. Ketcham, L. Christiansen, and G. Boitnott, Data report: permeability, resistivity, and X-ray computed tomography measurements in samples from the PACMANUS hydrothermal system, in Proc. ODP, Sci. Results, 193: College Station, TX (Ocean Drilling Program), edited by Barriga, F. J. A. S., R. A. Binns, D. J. Miller, and P. M. Herzig, 1–14, doi:10.2973/odp.proc.sr.193.205.2004, 2004.Google Scholar
- Kano, A., T. G. Ferdelman, T. Williams, J.-P. Henriet, T. Ishikawa, N. Kawagoe, C. Takashima, Y. Kakizaki, K. Abe, S. Sakai, E. L. Browning, X. Li, and The IODP Expedition 307 Scientists, Age constraints on the origin and growth history of a deep-water coral mound in NE Atlantic drilled in IODP Expedition 307, Geology, 35, 1051–1054, 2007.View ArticleGoogle Scholar
- McCullough, E. C., Photon attenuation in computed tomography, Med. Phys, 2, 307–320, 1975.View ArticleGoogle Scholar
- Mees, F., R. Swennen, M. Van Geet, and P. Jacobs, Applications of X-ray computed tomography in the geosciences, Geol. Soc., Lond., Special Publications, 215, 1–6, 2003.View ArticleGoogle Scholar
- ODP Information Technology and Data Services, ODP prime scientific data: Collection, archive, and quality, ODP Tech. Note, 37, doi:10.2973/odp.tn.37.2007, 2007.Google Scholar
- Orsi, T. H. and A. L. Anderson, Bulk density calibration for X-ray tomographic analyses of marine sediments, Geo-Mar. Lett., 19(4), 270–274, 1999.View ArticleGoogle Scholar
- Orsi, T. H., C. M. Edwards, and A. L. Anderson, X-ray computed tomography: A non-destructive method for quantitative analysis of sediment cores, J. Sediment. Res., A64, 690–693, 1994.View ArticleGoogle Scholar
- Pyrak-Nolte, L. J., C. D. Montemagno, and D. D. Nolte, Volumetric imaging of aperture distributions in connected fracture networks, Geophys. Res. Lett., 24, 2343–2346, 1997.View ArticleGoogle Scholar
- Shannon, P. M., P. D. W. Haughton, and D. Corcoran (eds.), The petroleum exploration of Ireland’s Offshore Basins, Geol. Soc., Lond., Special Publications, 188, 473 pp, 2001.Google Scholar
- Shimazaki, H. and S. Shinomoto, A method for selecting the bin size of a time histogram, Neural Comput., 19(6), 1503–1527, 2007.View ArticleGoogle Scholar
- Tanaka, A. and T. Nakano, Data report: three-dimensional observation and quantification of internal structure of sediment core from Challenger Mound area in the Porcupine Seabight off western Ireland using a medical X-ray CT, in Proc. IODP, 307, edited by Ferdelman, T. G., A. Kano, T. Williams, J.-P. Henriet, and the Expedition 307 Scientists, Washington, DC (Integrated Ocean Drilling Program Management International, Inc.), doi:10.2204/iodp.proc.307.209.2009, 2009.Google Scholar
- Titschack, J., M. Thierens, B. Dorschel, C. Schulbert, A. Freiwald, A. Kano, C. Takashima, N. Kawagoe, X. Li, and the IODP Expedition 307 scientific party, Carbonate budget of a cold-water coral mound (Challenger Mound, IODP Exp. 307), Mar. Geol., 259, 36–46, 2009.View ArticleGoogle Scholar
- Tivey, M. K. and S. Singh, Nondestructive imaging of fragile sea-floor vent deposit samples, Geology, 25, 931–934, 1997.View ArticleGoogle Scholar
- Wessel, P. and W. H. F. Smith, New, improved version of the Generic Mapping Tools released, Eos Trans. AGU, 79(47), 579, 1998.View ArticleGoogle Scholar
- Williams, T., A. Kano, T. G. Ferdelman, J.-P. Henriet et al., Cold-water coral mounds revealed, Eos Trans. AGU, 87(47), 525–526, 2006.View ArticleGoogle Scholar