- Open Access
NetPICOmag: A low-cost networked magnetometer and its applications
© The Society of Geomagnetism and Earth, Planetary and Space Sciences (SGEPSS); The Seismological Society of Japan; The Volcanological Society of Japan; The Geodetic Society of Japan; The Japanese Society for Planetary Sciences; TERRAPUB. 2012
- Received: 1 April 2011
- Accepted: 3 October 2011
- Published: 12 March 2012
NetPICOmag (NPM) is the culmination of a design effort to build a compact, low-cost, laboratory-grade, networked magnetometer designed for remote autonomous operation, suited for research and education. NPM allows wide placement of magnetometers sensitive enough to detect auroral activity and the daily variation, and is suitable for education projects and a range of geophysical applications. The use of networked microcontrollers and GPS timing is applicable to other small instruments for field or local deployment, and an onboard data logging capability has also been demonstrated. We illustrate the value of the placement of low-cost magnetometers to increase coverage in an area through the study of a Pc 5 pulsation event which took place on September 4, 2010. By combining results with those from auroral zone magnetometers supporting the THEMIS project, we find that the phase velocity of these morning sector pulsations was northward on the ground. The event took place under very quiet solar wind conditions, and credible mapping associates it with the inner magnetosphere. Another aspect beyond increasing areal coverage is increasing density of coverage, which becomes feasible with instruments of very low cost. We examine aspects of the April 5, 2010 space weather event which are possible to deduce from closely spaced magnetometers.
- Pc 5
Modern technology provides unique opportunities for researchers studying the magnetic effects of the aurora and the processes that drive electric currents in the Earth’s magnetosphere. Low-cost computers and instrumentation combined with widespread availability of wireless communication in remote regions has made the establishment of networked ground-based magnetometer stations technologically feasible. However, the cost of typical research-grade magnetometers is a significant obstacle to establishing high density, wide coverage magnetic detection networks consisting of many such instruments. We first discuss why such networks are needed.
The history of modern space physics is intimately linked with the ability to accurately measure magnetic fields (Russell, 1995). The diurnal variation, caused mainly by solar heating and easily measured with the instruments discussed below, was discovered through painstaking observation of compass needles by Hiorter in the early 1700s. In 1741, he was the first to make the connection between auroral and magnetic activity, a relation that our equipment would allow students in the auroral zone to see firsthand. As early as the 1830s, Gauss developed mathematical methods for the separation of internal and external fields, leading to the establishment of magnetic observatories that would allow the study of the fields from both sources.
A major technical step forward was the development of the fluxgate magnetometer in 1940 (Vacquier, 1993), which allowed the projection of the magnetic field along its axis to be accurately measured. Fluxgate magnetometers have become the workhorses of geomagnetism, both ground- and space-based. The sensor used in our equipment is commercially available and operates in a manner related to that of fluxgate magnetometers in which a core is driven into saturation.
Magnetic effects due to the daily variation and storms may be observed at any position on Earth, and the causative current systems are large and in general not highly structured as detected from observations at Earth-based locations. By contrast, auroral zone activity can feature structures on very small scales, and the structures which produce perturbations on the ground may be on the scale of auroral arcs. Although there is sometimes an argument made that it is not useful to place magnetometers closer together than the height of the ionosphere, our experience is that spatial scales of the perturbations on the ground can be smaller, notably in the vertical component.
Whether we really need to place magnetometers closer than 100 km apart is largely a moot point since existing networks have a vastly greater average spacing than this. Many of the fine-scale variations are associated with sub-storms, which we maintain cannot be adequately characterized from the ground with existing networks. Improving this situation within reasonable budget constraints has been a primary motivating factor in our development of low-cost magnetometers.
Ground based magnetometers are spatially integrating instruments that detect electric currents from sources both nearby and from great distances. A major source of the magnetic fields sensed by these magnetometers, especially in the auroral zones, lies in often narrow currents flowing roughly parallel to lines of constant magnetic latitude. These narrow electrojets are dynamic, changing in time and moving northward and southward as auroral activity develops and evolves. To follow these temporal and spatial developments requires measurements at half the scale imposed by the location of the current (i.e. 100 km) so that chains of magnetometers spaced every 50 km are needed over the latitude band in which the auroral electrojet occurs. This is true not only for quasi-static phenomena but also for traveling waves such as Pc 5 pulsations. Thus the exploration of auroral zone phenomena requires extensive deployment of magnetometers at potentially great cost. Current state-of-the-art magnetometers can cost many thousands of dollars, limiting the possible density and coverage of ground based magnetometer networks (Bredeson and Connors, 2004) and limiting scientific return. What is needed is a proverbial 500 dollar magnetometer.
This paper discusses the design and results from an ongoing project at Athabasca University whose goal has been to produce a low-cost research-grade magnetometer using digital fluxgate magnetometer technology. NetPICOmag (or NPM) is the latest development in Athabasca University’s ongoing low-cost magnetometer project (Connors et al., 2002). Such an instrument will permit the development of dense, wide-scale magnetometer observation networks, currently cost-prohibitive with available instrumentation. Increasing the spatial coverage of magnetic sensors across Northern Canada will improve research capabilities and lead to better understanding of the solar-terrestrial processes that produce auroral displays in the ionosphere.
A low-cost magnetometer must possess the following properties in order to be viable. First, it has to yield research-quality magnetic field measurements. This is typically considered to be the ability to detect magnetic perturbations as small as 1 nT sampled once per second (1 Hz). In comparison, the current model UCLA-built THEMIS Ground Based Observatory (GBO) fluxgate magnetometer can detect perturbations as small as 10 pT (Russell et al, 2008) sampled twice per second (2 Hz). A low-cost magnetometer must be easy to deploy and involve minimal maintenance. Practically speaking, installation involves burying and aligning the magnetic sensor probe, connecting it to the network, and powering it up. In addition, the unit must have network communication capability, whereby data can be streamed in near real-time over the Internet to a central data repository in order to operate autonomously. Finally, the magnetometer should be affordable so a wide group of users, which in addition to professional researchers could include radio amateurs and teachers, can participate in a large-scale magnetic observation network.
As of 2010, several NPMs have been distributed across Canada, particularly to sites in northern latitudes where magnetic fields show variation on the order or 100 nT or more due to substorm activity. They are currently sending data to our AUTUMN (Athabasca University THEMIS UCLA Magnetometer Network) magnetic data repository. The data is accessible at our public data portal (http://autumn.athabascau.ca), and at its sister site that is geared towards science education (http://auroranet.org). Stations are optimally installed in the non-winter months, allowing operators to bury instruments in the ground.
Mainly through fieldwork performed by the University of California (Berkeley and Los Angeles) and the University of Calgary, the NASA THEMIS mission emplaced two magnetic networks known as the Ground Based Observatory (GBO) and Geomagnetic Event Observation Network by Students (GEONS). The latter involves a dozen high schools across the United States as part the Education and Public Outreach (E/PO) effort of the current THEMIS mission to study the birth and evolution of sub-storms (Angelopoulos, 2008). Ten GEONS magnetometers were located at high schools within the United States, contributing magnetic data for the mission as well as educating students and the public about geomagnetism, space weather, the aurora and the THEMIS mission (Peticolas et al., 2008). Both GEONS and the THEMIS GBO network use the aforementioned UCLA THEMIS-GBO flux-gate magnetometer, each supported by a networked PC that transmits data in real-time data and also archives it locally to hard drive.
The GBO network couples the magnetometers with im-agers, and forms an irregular grid of networked stations stretching from Newfoundland to Alaska. In general, GBO site locations were chosen for optimal optical coverage. The desired structure of the NPM instrument network is patterned after the THEMIS magnetometer network in which a network of autonomous, Internet-enabled instruments continuously feed data to a central data repository that can be browsed using its web-based data gateway. However, we have identified use of a support computer as a failure point (as well as expense and power draw) in networks based on GBO magnetometers, such as our own AUTUMN network. In addition, the need to have much greater spatial coverage forced us to consider low cost as an essential factor in our design.
We proceed to discuss NPM’s components and method of construction, performance compared to current UCLA flux-gate magnetometers, and possible deployment strategies.
2.1 Instrument design
2.1.1 Magnetic sensors
The heart of the NPM is its three Speake and Company FGM-3 digital fluxgate magnetometers (Speake & Co. Llanfapley, 2010). These sensors use a proprietary design based on traditional fluxgate magnetometer technology. We have found these sensors to be sufficiently sensitive for use in geomagnetic research, but they are highly sensitive to temperature change. After considerable effort to resolve this issue, we settled on a design similar to the UCLA THEMIS-GBO magnetometers, which places the sensors in a tube buried vertically in the ground to provide thermal inertia. Three sensors located in the bottom of the tube are orthogonally oriented to measure the geomagnetic field strength in terms of local magnetic coordinates where X M is the component aligned with magnetic north, Y M with magnetic east, and Z M points vertically downward.
The FGM-3 emits a TTL-conformant (0-5V DC) digital pulse stream whose frequency is non-linearly proportional to the strength of the surrounding magnetic field, but approximately linear for small perturbations. A change in 1 Hz translates approximately into a 1 nT difference. The FGM-3h variant offers an approximate 2.5-times boost in sensitivity at the expense of dynamic range, allowing the sensor to detect a 400 picotesla (pT) change in the magnetic field. The NPM uses the less sensitive FGM-3 to measure the relatively strong B Z component (as seen in high latitudes), while FGM-3h sensors are used for the B X and B Y components.
As mentioned, the FGM series of sensors are sensitive to ambient temperature. Therefore, it is crucial that FGM sensors be placed in thermally stable environments, preferably underground. Thermal isolation of the magnetic sensor probe has proven to be the best solution to counteract thermal biasing of the magnetic field data. Seasonal temperature changes appear in the data when viewed over a large time span. Baselining the data adequately compensates for this effect for the purposes of auroral research. This is achieved by taking the average of magnetic data from a recent magnetic quiet period and subtracting it from the immediate data, yielding a time series depicting magnetic field perturbations relative to the seasonal quiescent magnetic field. Similar correction is required even for observatory-grade magnetic detectors.
The NPM housing isa1mby100 mm (sold as 4-inch) diameter ABS pipe. The top end (appearing to the left in Fig. 1) has a removable weatherproof threaded cap, while the bottom end appearing on the right is sealed. The tube is buried vertically in the ground so that approximately 15 cm of the top end of the tube lies exposed above ground. This allows the internal GPS antenna (the black disk to the left, pictured in Fig. 1) to receive adequate signal for generating a one-pulse-per-second (1 PPS) heartbeat or master timing signal. The top of the NPM housing is painted white to reflect solar radiation in order to reduce internal heat build-up. A polyurethane foam plug roughly 8 cm thick provides additional thermal isolation for the magnetic sensors located near the bottom of the tube.
A wood mounting board holding the electronics and the magnetic sensors sits firmly inside the tube when the top cap is screwed shut, and can be easily slid out the tube when the cap is removed. A weatherproof RJ-45 feed-through connector passing through to the top wall of the tube allows a standard Cat5 Ethernet cable to plug into the side of the NPM unit, allowing a secure, moisture-resistant connection.
The NPM design is considerably simpler and more compact than a similar research-grade magnetometer, such as the THEMIS GBO fluxgate magnetometer. The THEMIS GBO magnetometers that were distributed as part of the NASA THEMIS mission consisted of multiple components: 1.1 m long fluxgate magnetometer sensor tube, AC-powered magnetometer electronics unit, GPS antenna and receiver card, desktop PC and display, uninterruptable power supply (UPS), remote booting device, assorted cables, and data collection software (Russell et al, 2008). NPM was intentionally designed to encapsulate these components into a single unit in order to simplify installation. The operator of an NPM unit only needs to bury the instrument and provide power and Internet access. We chose to eliminate the desktop PC computer entirely because we felt was a major point of failure in terms of software and hardware reliability.
2.1.2 Frequency counters
The FGM-3 and FGM-3h sensors both produce TTL pulse streams with frequencies ranging from the 10s to 100s of kHz. We have found that the 8-bit “PIC” series of microcontrollers from Microchip, Inc. are ideal high-speed pulse counters, because, unlike many similar microcontrollers, PIC microcontrollers allow external clocking of some of their internal 16-bit registers, which can thus act as counters. The NPM uses twin PIC18F252 microcontrollers (Microchip Technology Inc., 2006), each having two externally clocked 16-bit counters. Because the output frequency of the magnetic sensors is sufficiently high to overflow the counters over the course of one second, frequency is measured by reading the stored counts added to the counter range multiplied by the number of timer overflows that occurred over the course of 1 second. The PIC counters return the resulting frequency measurement over a 9600-baud TTL-level serial data stream, which is easily read by the control microcontroller. As each PIC is able to count two pulse streams, one of the PICs returns the frequencies of sensors X and Y, while the second PIC returns the frequency of sensor Z, leaving a spare channel.
In order to compute the pulse frequency, an accurate 1-second time base must be provided as the master clock or heartbeat. The one pulse per second (1 PPS) signal from a GPS board is easy to obtain, as most OEM GPS boards provide this signal, which is accurate to ±1μS, making it an ideal time base for measuring the frequency of the magnetic sensor output signal. The master clock is implemented through an interrupt, with counter reset and transmission of frequency data as part of the interrupt service routine executed by the CPU (and thus not affecting the ability of the timing registers to count). Time and date information aligned to the leading edge of the 1 PPS signal is fed to the controlling microcontroller (MCU) module over a TTL serial link, and is encoded in industry-standard NMEA 0183 format (National Marine Electronics Association, 2002). We selected the Garmin GPS 18 LVC GPS engine (Garmin International Inc., 2005) for its compactness and simplicity. It is a 61 mm diameter by 19 mm thick disk containing an integrated receiver and antenna attached toa5m power/communication cable. The unit was configured to emit a 1 PPS pulse and the NMEA GPRMC sentence, thus providing the 1 Hz timing pulse and time stamps. Configuration settings are stored in the GPS unit’s non-volatile memory.
NPM uses a Rabbit Semiconductor RCM4010 8-bit networked microcontroller module containing a 59.98 MHz Rabbit 4000 CPU with 512 kB of SRAM and 512 kB of non-volatile FLASH program storage (Digi International Inc., 2010). The pulse counting PICs and additional electronics are integrated on a custom-designed printed circuit board (PCB) that provides on-board regulated 3.3 and 5 DC voltage, plug in dual inline package (DIP) sockets for mounting DIP components, and a plug-in header on which to mount the Rabbit RCM4010 MCU core module.
Previous experience with Rabbit 2000 microcontroller modules in networked applications (Schofield and Naylor, 2007), showing them to be capable of reliable internetwork communication, led to the decision to port the project from an initial design based on a Microchip “dsPIC” microcontroller board to the Rabbit platform. In the course of a few weeks, we were able to convert the needed part of PIC-based source code to Rabbit Semiconductor’s Dynamic C (version 10.23).
The NPM control software centers around an interrupt service routine (ISR) triggered by a 1 PPS signal from the GPS. The 1 PPS signal alerts the software to read the current magnetometer sensor frequencies from the PIC pulse counters. Then, the GPS unit’s date-time string formatted as a NMEA 0183 GPRMC sentence is read by the Rabbit MCU. Frequency and time information are buffered until ten samples are collected and then transmitted to a remote data collection server over the UDP datagram network socket (Postel, 1980). 10 second data blocks are 440 bytes in size. This size was selected in order to avoid fragmenting the data across several UDP packets and to use the space provided in a default 512 byte UDP packet in an efficient manner.
The UDP protocol used for data transmission is relatively simple and has little overhead, thus it lends itself to use on small embedded appliances such as NPM. However, UDP transmission does not guarantee delivery of transmitted packets, nor the order in which they are received. In situations where an NPM unit must send data that must cross multiple networks (internetworking) as opposed to a local area network (LAN) environment, packet loss is a distinct possibility, particularly where network quality of service is questionable. For this reason, onboard logging onto a nonvolatile flash memory device is desirable for NPM units that require an assured record of their data. A separate NPM unit based around the Rabbit 4300 core module that uses mini SD memory was successfully tested. Buffered data is written to the mini SD card formatted with a FAT file system. This is done only every 60 seconds in order to minimize write cycles, thus extending the lifespan of the storage media. An additional advantage of this memory-based storage is that in principle a device could be developed that operated without being attached to a network, with an operator retrieving the data manually as needed or dictated by the storage capacity.
To simplify installation, the NPM is a single integrated unit containing the GPS, magnetic sensors and control electronics. It receives power and network over a Cat5-E Ethernet cable configured as a Power-over-Ethernet (PoE) cable, where two otherwise unused twisted pairs carry DC voltage and ground. A PoE cable can carry power and communications up to 100 m, nominally at data rates of 100 Mbit/s. In tests, we achieved best results with PoE cable lengths up to 75 m, beyond which Ethernet speeds needed to be reduced to 10 Mbit/s.
Since the unit is small, costs of network deployment can be minimized by sending it to the cooperating install site rather than making a field trip. NPM was designed with simplicity of installation in mind, requiring only a few manual tools. A posthole digger is used to make a∼1 m deep hole into which the unit is placed vertically and secured with sand. The NPM must be placed in an area free of pedestrian or vehicular traffic, far from large magnetic objects such as electric motors and motor vehicles. The top of the NPM tube remains above ground and must have a clear view of the sky so that GPS signals may be received. Once the NPM unit is buried in an acceptable location and aligned to magnetic north (easily done using a handheld compass), the operator connects power and network into a D-Link DWL-P200 PoE base unit (D-Link Corporation, 2004) and runs a Cat5-E Ethernet cable from the base unit to the NPM. The microcontroller is configured in firmware to search for its network configuration using the dynamic host configuration protocol (DHCP).
When the magnetometer controller is powered up, data collection begins upon receipt of the 1 PPS signal. This occurs when the GPS unit calculates a fix on the minimum required number of satellites. On a cold start, this can take about a minute with the particular Garmin GPS module used by NPM. Afterwards, packets of ten data points are transmitted to our data logging server every ten seconds over a pre-defined range of network ports. It is necessary that the magnetometer site’s firewall, if one is in use, be configured to allow outgoing network traffic over these ports.
Instrument tests that started in early 2008 are still ongoing at the Athabasca University Geophysical Observatory (AUGO). AUGO is located on the Athabasca University campus, on the outskirts of the town of Athabasca, Alberta, geodetic latitude 54.71 north, 113.31 west (CGM coordinates: 61.88 north, 302.22 longitude). This falls near the southern boundary of the northern auroral zone, the region where auroral activity is most visible in the northern hemisphere. Over North America, this latitudinally confined belt extends across Alaska, Western Canada, Northern Quebec and Ontario. At auroral latitudes, the B X and B Z components tend to dominate the signal, being driven by the east/west electrojet in the ionosphere. At subauroral latitudes, the B Y component predominates, and is the result of field aligned currents.
Both magnetometers sample the magnetic field at 1 Hz and are synchronized to GPS timing, so no time base correction on the data is required. In order to correct for visible curvature in the original NPM data, second order polynomial curves were fitted to the NPM data. Subtracting the fit curve from the original data yields a flatter data set. Given that the instrument is a technology demonstrator, the sensor mounting hardware consists of a wooden plank with orthogonally oriented grooves for the sensors to sit in and fastened with plastic cable ties. We attribute some very small differences to sensor misalignment. In principle, a rotation of axes could be done mathematically to correct this. What we regard as more important is that small perturbations show a high degree of similarity between the NPM and SMALL instruments.
4.1 Ground observations
4.2 Mapping to the magnetosphere
Due to quiet conditions prevailing, and the small amplitude of the pulsations compared to the background magnetic field, it is reasonable to expect that empirical magnetospheric models would provide a good indication of the mapping of the pulsations into space. Mappings from the ground stations discussed using the T89 model (Tsyganenko, 1989) are shown in Fig. 9. This model is based on averaging magnetic field observations made in space by several spacecraft prior to 1989. Also shown is the Petrinec-Russell (1996) magnetosphere boundary, similarly based on empirical fits to readily identifiable magnetopause boundary crossings by spacecraft. The IMF B Z =-1nT and P = 0.7 nPa parameters used are not unusual in the solar wind, so that the plotted boundary is expected to conform well to that during this event. Also shown are the approximate range of plasmapause locations for the dawn sector for quiet conditions (O’Brien and Moldwin, 2003). Once more, this is derived from empirical fits to a readily identifiable boundary, but with a smaller dataset having been used than in the magnetic field and boundary models, the range of possible values is larger. It is seen that the ATH and GPR stations map to near the plasmapause, while the magnetopause is remote from the mapping location and in the direction toward which the pulsations propagate. Data from other AUTUMN chain stations south of ATH, such as RED, indicate a rapidly decreasing amplitude of the pulsations at those locations, which would map to near or inside the plasmapause. This pulsation event appears to have had waves originating at the plasmapause, closely associated with the time during which the IMF B Z was negative. This suggests an association with reconnection-induced changes in the inner magnetosphere. The present study is not able to indicate more. However, it shows that the deployment of our low-cost magnetometers allowed the identification of a type of pulsational behavior of which we were previously unaware, and apparently not yet documented.
We suggest that further study of such events is merited. Further deployment of large arrays, enabled by low-cost magnetometers, would be an effective way of furthering studies of such pulsations and other phenomena.
The question of the optimal spacing of magnetometers was a moot point when the instruments were relatively expensive and difficult to install. It has been widely claimed that a spacing of less than approximately the height of the ionosphere is not useful since magnetic signals should not vary on much smaller scales. In general the spacing of existing magnetometers is much greater than the ca. 100 km separation this criterion would imply, and the matter has not been put to the test. Here we describe an auroral event in which a small magnetometer spacing revealed significant gradients in magnetic field.
5.1 Comments on density of coverage
In this example, the spacing of stations near the auroral oval was 2 to 3 degrees, i.e. roughly 250 km. The model appears to have captured the essential features of the electrojet situation, especially given the good match to data. However, the maximum in X due to the eastward electrojet is represented by only one point, and the gradient near the electrojet interface only by two points. Clearly, extra stations would not be redundant, and could sharpen the resolution in the interface region.
In specific auroral events, there would be two advantages to wide and dense coverage which low-cost magnetometers allow. Stations filling an area allow detection of events where they might not have been detected without that coverage. Runov et al. (2011) examine the timing of a dipolarization front observed by the THEMIS spacecraft relative to a minor current intensification observed from the ground. The latter was extremely localized and detected clearly at only one station (Yellowknife) which was not part of the THEMIS ground network. Proper study of the event benefitted from the de facto increase in density by adding this station. Placement of more magnetic instruments enhances the precision with which timing studies can be done. The second advantage is that details of the auroral currents can be discussed with higher spatial resolution if densely placed stations exist where this resolution is needed.
5.2 Localized auroral currents
Since we are in the evaluation phase of our project, we do not yet have a dense network deployed. However, the major auroral event of April 5, 2010 (Connors et al., 2011) had large signals detected in western Canada, including at the NPM site at Grande Prairie, Alberta. The STEP Polar Network site at Fort St. John, British Columbia, is 166 km NW of Grande Prairie (see Fig. 9) and houses a standard Narod fluxgate magnetometer. Compared to most other existing site pairs, these sites are relatively close together and suited to investigating whether closely spaced sites are useful during an auroral event. A comparison of the differences in observed signals during this large event is best based on cross-calibration during the event itself. An auroral camera was operating at Athabasca (AUGO observatory, equivalent to the magnetic station ATH) during the event with FSJ and GPR on the edge of the field of view. We can thus compare inferences based on both optical and magnetic data.
5.3 Linearity of sensors during a large auroral event
The main Athabasca observatory (AUGO) has a UCLA SMALL fluxgate magnetometer whose designation is ATH, and at times has housed several NPMs under test and calibration. At the time of the April 5, 2010 event, one NPM was in operation near the ATH fluxgate. Due to the very large perturbations during this event, we can determine whether the NPM sensors deviate significantly from linearity during the largest expected magnetic events. In addition, we can add to the above discussion about field-aligned currents.
One other aspect of examining the ATH data is that the positive Y noted to have been present at GPR near 9 UT was not present at ATH. From the auroral imaging, the upward current region was inferred to be west of ATH and would not have caused a Y component perturbation in the period prior to 09:10 UT. A field-aligned current effect similar to that observed at GPR is absent. Since the SMALL magnetometer is known from years of use to be well aligned with magnetic coordinates, its indication of a negative Y perturbation at this time instead indicates downward field aligned current north of ATH. With the onset at 09:09 UT, this effect became more pronounced, with a large increase in the magnitude of the (negative) Y component. This was due to the downward field-aligned current of the substorm current wedge (McPherron et al., 1973) whose westward electrojet lay northwest of Athabasca as evidenced by the enhanced (negative) X perturbation at CMO at this time (Fig. 11).
5.4 Implications from April 5, 2010 event
This section has given only an indication of what can be inferred from closely spaced stations in an auroral event. It is not intended as an exhaustive analysis of the ground signatures from the April 5,2010 event. The implication is that closely spaced stations near intense aurora (as observed to be present in the region using the Athabasca all-sky camera) may be used to infer details of the current flow, a conclusion already reached many years ago by Baumjohann and Opgenoorth (1983). Having wider coverage enhances the chances of having data at interesting places; having denser coverage enhances the ability to interpret the data. We further note that the Scandinavian Magnetic Array (Küppers et al., 1979) had relatively dense coverage on the subcontinental scale of Scandinavia but that the ability to efficiently manage digital data has since greatly improved, enhancing the value of digital magnetometers (such as ours) which may now be used to implement such arrays.
We have shown examples in which event interpretation was enhanced by wider and denser coverage by magnetometers. For phenomena in which spacing of 250 km would be useful, this could be attained in many regions of northern Canada where instruments could be sent by post, and where Internet connectivity is available, Canada’s10 million square kilometers could be covered by about 160 instruments. The cost of implementing this, assuming local participation on a volunteer basis, would be only about $80,000 in instrumentation. Our present Grande Prairie instrument was delivered and then installed by a local colleague, and one could envisage participation from local residents interested in science and/or education. The new technology we have used would allow extensive data coverage at extremely low cost. Much new insight could be obtained with smaller scale dense arrays. A 13 by 13 array of 169 magnetometers could cover a region of 600 by 600 km and allow observers to watch the onset of events and their phase and group propagation, providing a giant “TV screen” for watching ionospheric currents and magnetic pulsations. The deployment of such a system would be practical in rural regions under the auroral and subauroral ionosphere, such as are found in some regions of Canada and other northerly countries.
Tests demonstrating the performance of a networked, 3-axis, 1-sps, digital pulse counting fluxgate magnetometer have produced data that show close correlation with data collected from a research-grade UCLA-built SMALL 1 Hz fluxgate magnetometer. With a material cost of approximately USD $500, the NPM offers a high cost to performance ratio with 1 nT resolution and 1 Hz data sampling of three axes, which is considered the minimum standard for a research-grade instrument. Many projects could take advantage of the NPM, either as a teaching tool that allows students to work with real, locally-produced scientific data or in the development of large scale magnetic sensor networks that would be cost prohibitive using current instrumentation.
We have also demonstrated a generally usable microcontroller-based platform which could transmit data to either a local or remote archiving computer. Deployed stations have operated continuously and reliably for several years. This suggests that the low-cost, low-power and reliable approach we have implemented for magnetometer data gathering may be widely applicable in gathering any data with a relatively slow sampling rate. With some minor technical improvements and minimal funding, and by approaching potential cooperating bodies at the appropriate administrative level, we hope to be able to meet the goal of having a much denser network of magnetometers in Canada.
Data for Fig. 10 was obtained from the Principal Investigator of the CANOPUS magnetic array at the time, Gordon Rostoker. Fort St. John magnetometer operations are conducted in cooperation with Kanji Hayashi (University of Tokyo, retired) with the support of Neil Thompson. The AUTUMN Red Deer magnetometer is operated through the cooperation of Dr. B. Martin (King’s University College), and Mrs. E. Martin. USGS CMO data was obtained through Intermagnet. Mapping of auroral camera data was done with the help of Emma Spanswick. Thanks to Ali M. Al-Asadi (Grande Prairie Regional College) for obtaining and installing a NPM (GPR). We acknowledge CDAWeb for access to data, J. H. King and N. Papatashvilli for OMNI data, and the Government of Canada through the Canadian Space Agency and NRCan for magnetometer data (including partial funding of auroranet.org development). NPM instruments were developed using support associated with a Canada Research Chair. Implementation of cross-correlation analysis was aided by code placed online by P. Bourke at http://local.wasp.uwa.edu.au. Special thanks to Jill Mattock at Paul Rowe Jr./Sr. High School and the information technology staff at the Peace River School District for their installation and testing of the first school-installed NPM.
- Angelopoulos, V., The THEMIS mission, Space Sci. Rev., 141(1/4), 5–34, 2008.View ArticleGoogle Scholar
- Baumjohann, W. and H. J. Opgenoorth, Electric fields and currents associated with active aurora, in Magnetospheric Currents, edited by T. A. Potemra, 357 pp., AGU, Washington, 1983.Google Scholar
- Baumjohann, W. and R. A. Treumann, Basic Space Plasma Physics, 329 pp., Imperial College Press, London, 1997.Google Scholar
- Box, G. E. P. and G. M. Jenkins, Time Series Analysis (Revised Edition), 575 pp., Holden-Day, Oakland, 1976.Google Scholar
- Bredeson, C. and M. Connors, Enhancing the study of the aurora using low-cost magnetometers, Can. Undergrad. Phys. J., 2(2), 9–13, 2004.Google Scholar
- Connors, M., J. Ponto, and M. Foote, Assessment of pulse counting magnetometer technology for solar-terrestrial environment monitoring, AGU, Fall Meeting 2002, SM21B-0544, 2002.Google Scholar
- Connors, M., C. T. Russell, and V. Angelopoulos, Magnetic flux transfer in the 5 April 2010 Galaxy 15 substorm: An unprecedented observation, Ann. Geophys., 29, 619–622, doi:10.5194/angeo-29-619-2011, 2011.View ArticleGoogle Scholar
- Digi International Inc., RabbitCore RCM4000 C-Programmable Analog Core Module With Ethernet: User’s Manual, http://ftp1.digi.com/support/documentation/019-0157_J.pdf, 2010.
- D-Link Corporation, DWL-P200 Data Shet, ftp://ftp10.dlink.com/pdfs/products/DWL-P200/DWL-P200_ds.pdf, 2004.
- Frigo, M. and S. G. Johnson, The design and implementation of FFTW3, Proc. IEEE, 93, 216–231, 2005.View ArticleGoogle Scholar
- Fukushima, N., Ground magnetic effect of field-aligned currents connected with ionospheric currents, Geophys. Res. Lab. Rep., UT-GRL-76-01, University of Tokyo, 1976.Google Scholar
- Gao, Y. F.,et al., Sino-magnetic array at low latitudes (SMALL) including initial results from the sister sites in the United States, Adv. Space Res., 25(7/8), 1343–1351, 2000View ArticleGoogle Scholar
- Garmin International Inc., GPS 18 Technical Specifications, http://www8.gamin.com/manuals/425_TechnicalSpecification.pdf, 2005.
- Kisabeth, J., On calculating magnetic and vector potential fields due to large-scale magnetospheric current systems and induced currents in an infinitely conducting earth, in Quantitative Modeling of Magnetospheric Processes, edited by W. P. Olson, 655 pp., AGU, Washington, 1979Google Scholar
- Kivelson, M. G., ULF waves from the ionosphere to the outer planets, in Magnetospheric ULF Waves, edited by K. Takahashi et al., 359 pp, AGU, Washington, 2006.Google Scholar
- Küppers, F., W. Untiedt, W. Baumjohann, K. Lange, and A. G. Jones, A two-dimensional magnetometer array for ground-based observations of auroral zone electric currents during the international magnetospheric study (IMS), J. Geophys, 46, 429–450, 1Google Scholar
- McPherron, R. L., C. T. Russell, and M. P. Aubry, Satellite studies of magnetospheric substorms on August 15, 1968. 9. Phenomenological model for substorms, J. Geophys. Res., 78, 3131–3149, 1973.View ArticleGoogle Scholar
- Microchip Technology Inc., Microchip PIC18FXX2 Data Sheet, http://ww1.microchip.com/downloads/en/DeviceDoc/39564c.pdf, 2006.
- National Marine Electronics Association (U.S.), NMEA 0183 Standard for Interfacing Marine Electronic Devices, NMEA, New Bern, NC, 2002.Google Scholar
- O’Brien, T. P. and M. B. Moldwin, Empirical plasmapause models from magnetic indices, Geophys. Res. Lett., 30, doi:10.1029/2002GL016007, 2003.
- Peticolas, L. M., et al., The time history of events and macroscale interactions during substorms (THEMIS) education and outreach (E/PO) program, Space Sa. Rev., 141(1/4), 557–583, 2008.View ArticleGoogle Scholar
- Petrinec, S. M. and C. T. Russell, Near-Earth magnetotail shape and size as determined from the magnetopause flaring angle, J. Geophys. Res., 101, 137–152, 1996.View ArticleGoogle Scholar
- Postel, J., User datagram protocol, RFC 768, IETF, http://tools.ietf.org/html/rfc768, 1980.
- Press, W. H., S. A. Teukolsky, W. T. Vetterling, and B. P. Flannery, Numerical Recipes in C, 994 pp., Cambridge, New York, 1992.Google Scholar
- Runov, A., V. Angelopoulos, M. Sitnov, V. A. Sergeev, R. Nakamura, Y. Nishimura, H. U. Frey, J. P. McFadden, D. Larson, J. Bonnell, K.-H. Glassmeier, U. Auster, M. Connors, C. T. Russell, and H. J. Singer, Dipolarization fronts in the magnetotail plasma sheet, Planet. Space Sa., 59, 517–525, doi:10.1016/j.pss.2010.06.006, 2011.View ArticleGoogle Scholar
- Russell, C. T., A brief history of solar-terrestrial physics, in Introduction to Space Physics, edited by M. G. Kivelson and C. T. Russell, 568 pp., Cambridge University Press, Cambridge, 1995.Google Scholar
- Russell, C. T. et al., THEMIS ground-based magnetometers, Space Sci. Rev, 141(1/4), 389–412, 2008.View ArticleGoogle Scholar
- Samson, J. C., Geomagnetic pulsations and plasma waves in the earth’s magnetosphere, in Geomagnetism, v. 4, edited by J. A. Jacobs, 832 pp., Academic Press, New York, 1991. Schofield, I. S. and D. A. Naylor, Control and communications system for remote operation of an infrared radiometer, Publ. Astron. Soc. Pacific, 119, 661–668, 2007.Google Scholar
- Singer, H. J., Multisatellite observations ofresonanthydromagnetic waves, Planet. Space Sci., 30, 1209–1218, 1982.View ArticleGoogle Scholar
- Speake & Co. Llanfapley., FGM-series Magnetic Field Sensors (datasheet), http://www.speakesensors.com, 2010.
- Tsyganenko, N., A magnetospheric magnetic field model with a warped current sheet, Planet. Space Sci., 37, 5–20, 1989.View ArticleGoogle Scholar
- Vacquier, V., Many jobs, Ann. Rev. Earth Planet. Sci., 2, 1–17, 1993.View ArticleGoogle Scholar